Thursday, May 2, 2013
In spite of already being a working device, there were aspects that like I explained in the previous post, required some attention. One of the major fixes was the voltage drift. This was caused by the 5 K potentiometer used to set the voltage end point. It was too large (the sweet spot of my setup was around 500 Ohms), which besides making it difficult to adjust, caused temperature induced resistance drift to be too large, causing an increase of around 200 mV from the initial voltage once the temperature settled.